Message 4 of 4. SCPI query to an instrument to get its identification string. Also, FYI, it’s usually better to post questions about instrument control to the Instrument Control forum here:. Legal Privacy Terms Trademark Acknowledgements. See search results instead: Here is a list of tested printers and other printers that are likely to work Computer Software Current Version:
|Date Added:||15 April 2013|
|File Size:||68.92 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
A Delay measurement is made between channels 1 and 2 within each segment and the measurement value from each segment is listed.
Oscilloscopes | Keysight Community
I now want to This sample program captures and saves a screen image from the InfiniiVilsion X-Series oscilloscope. Is there a way to perform thi I haven’t been able to find any know issues with this driver. I had a B bw 50G installed on D. I’ve searched the forums but no one with my particular model of scope has posted on this problem yet. Please tell me how to fix this problem.
With Series you can make floating or differential measurements using a differential probe. This script shows how to properly initialize a Keysight InfiniiVision or InfiniiVision X-Series oscilloscope and synchronize an acquisition using the blocking: Although the sample rate cannot be directly set on any InfiniiVision oscilloscope, it can be set indirectly labviw adjusting the oscilloscopes time per division setting.
Not on the front panel, but you can do it with remote programming commands. Our B oscilloscope hard drive seems to be dead.
Agilent 6000 Series Oscilloscope
See search results instead: I do have the latest of all the drivers that I need. You need to upgrade your oscilloscope to the latest firmware Get Waveform Data ElegantVISA library, Python This script shows how to pull the waveform data from the analog channels for already acquired data and save it to a computer.
Then, I tried to use preamble block to do so, but it also didn’t work. Within the “Save” menu, press Settings and select the max number of points.
ag waveform capture errors – query UNTERMINATED – Discussion Forums – National Instruments
BenchVue Software Platform BenchVue software offers multiple instrument measurement visibility and data capture from a PC or labciew device with no programming. If you have your order number and certificate number, you can obtain the license key for your product at We need to replace the disk with another one old 2,5″ laptop drive and install an image of the disk which would recover the oscilloscope into the factory default It labgiew that the ID query failed every time I tried to run it.
Dsl6054a able to run: It can also get time tags and produce an averaged segment for each analog channel. BenchVue software offers multiple instrument measurement visibility and data capture from a PC or mobile device with no programming.
There was a fix posted for different agilent drivers with a similar error, but I don’t think I have the right software to modify the IVI driver.
Keysight Technologies / Agilent Technologies DSO6054A
Can the Series oscilloscopes make floating measurements? This sample program assumes the oscilloscope has been set to capture N segments.
Do the mean, min, and max values of the InfiniiVision oscilloscopes measurement statistics refer to one signal on the screen or all previously capt When I disabled the query, dso60054a VI returned only 0 when it should have been a sine wave. Like most other digitizing oscilloscopes, the new Series is not designed to offer floating measurement by itself.
Computer Software Current Version: